Magnetic Resonance Imaging (MRI) Safety Information for Zimmer Implants
Magnetic Resonance Imaging (MRI) is a commonly accepted and widely
used diagnostic medical procedure. It is often safe to perform MRI on
an individual that has an orthopaedic implant device. The main issues
affecting the safety of passive implants (medical devices that serve
their function without the supply of power) in the MR environment
involve magnetically induced displacement force and torque and radio
frequency (RF) induced heating. The MR static field induces
displacement forces and torques on ferromagnetic materials. However,
both ferromagnetic and non-ferromagnetic metallic devices of certain
geometries may experience heating caused by interactions with the RF
field. Of secondary concern is the possibility of image artifacts that
can compromise image quality.
Zimmer metallic implants are manufactured using one or more of the following non-ferromagnetic materials: commercially pure titanium (CP Titanium), Ti-6Al-4V alloy, Ti-6Al-7Nb alloy, several Co-Cr alloys (ASTM F75, F562, and F90), tantalum (Trabecular Metal™ Material), and the following implant grade stainless steels: 316L, REX 734, 22-13-5 and Biodur 108. Zimmer’s internal testing has revealed that although each metallic material exhibits a small but measurable magnetic attraction in the 1.5 Tesla and 3.0 Tesla environments, the maximum magnetic force exerted on a device (stainless steel) is less than 25% of the force exerted on the device due to gravity1. None of the metallic materials exhibited any torque movement in 1.5 Tesla and 3.0 Tesla MR environments. Therefore, no movement or deflection of Zimmer devices manufactured from the aforementioned metallic materials is expected in 1.5 Tesla and 3.0 Tesla MR environments.
The polymer (plastic) and ceramic materials used in the manufacture of some Zimmer implants are non-metallic and non-ferromagnetic and pose no risk of movement or deflection due to exposure to the MR environment.
In regards to RF induced heating, one recent publication states “…..to date, there has been no report of a patient being seriously injured as a result of excessive heat that developed in a "passive" metallic implant or device. However, heating is potentially problematic for implants that have an elongated shape or those that form a conducting loop of a certain diameter2.” The RF induced heating of Zimmer implants is currently being tested and the results will be made available upon completion.
Patients should note that there are several different manufacturers and generations of MRI systems available, and Zimmer cannot make any claims regarding the safety of Zimmer implants and devices with any specific MR system.
If more information is needed, please call Michael E. Hawkins, Ph.D., 800-613-6131, extension 14624 or David M. Miller, Ph.D., 800-613-6131, extension 18381. If calling from outside the US, please call +1.574.267.6131.
Zimmer Research Report_WA_2179_10_Rev.1
F. G. Shellock, “Reference Manual for Magnetic Resonance Safety, Implants, and Devices”, 2008 edition, pg:136
All content herein is protected by copyright, trademarks and other intellectual property rights, as applicable, owned by or licensed to Zimmer Biomet or its affiliates unless otherwise indicated, and must not be redistributed, duplicated or disclosed, in whole or in part, without the express written consent of Zimmer Biomet.
This material is intended for health care professionals. Distribution to any other recipient is prohibited.
For product information, including indications, contraindications, warnings, precautions, potential adverse effects and patient counseling information, see the package insert or contact your local representative; search this website for additional product information. To obtain a copy of the current Instructions for Use (IFU) for full prescribing and risk information, please visit labeling.zimmerbiomet.com or call 1-800-348-2759, press 4 for 411 Technical Support.